|Year : 2018 | Volume
| Issue : 4 | Page : 340-344
Simulated damage of two implant debridement methods: Nonsurgical approach with Teflon and stainless steel hand scalers
João Paulo Mendes Tribst, Amanda Maria de Oliveira Dal Piva, Dimas Renno de Lima, Alexandre Luiz Souto Borges, Marco Antonio Bottino
Department of Dental Materials and Prosthodontics, São Paulo State University (Unesp), Institute of Science and Technology, São José Dos Campos, São Paulo, Brazil
|Date of Submission||27-Mar-2018|
|Date of Acceptance||07-May-2018|
|Date of Web Publication||20-Jul-2018|
Amanda Maria de Oliveira Dal Piva
Department of Dental Materials and Prosthodontics, São Paulo State University (Unesp), Institute of Science and Technology, 777 Avenida Eng. Francisco José Longo, São José Dos Campos, SP 12245000
Source of Support: None, Conflict of Interest: None
| Abstract|| |
Background: Mechanical scaling is the most common treatment of periodontal and peri-implant tissue infections. Aims: This study aimed to evaluate the effect of mechanical prophylactic therapy on the residual stresses in the implant and hand scaler. Settings and Design: For finite-element analysis, an implant-supported prosthesis was created using modeling software with 3 mm of exposed threads. For simulation of a prophylactic mechanical debridement, the active face of the shank was disposed of in contact with the last thread exposed at a 90° angle. Materials and Methods: In the analysis software, the contacts were defined as rough between the instrument and the implant. The cortical bone was fixed and a load of 10 N was applied to the instrument cable. Two simulations were performed according to the instrument material: stainless steel or Teflon. Von-Mises results were obtained. Statistical Analysis Used: No statistical test was used, but, the 500 higher stress peaks in the implant and in the instrument were analyzed for qualitative comparison. Results: Mechanical prophylactic therapy generates higher residual stress on the implant with a stainless steel instrument. There was no difference between the materials for the active tip of the instrument, and the active portion of the shank was the region which concentrated more stress. Conclusions: It is suggested that hand scalers in Teflon are less damaging to the implant, but more susceptible to deformation and possible early failures.
Keywords: Cleaning instruments, dental/oral implants, hand instruments, implant surfaces, peri-implantitis
|How to cite this article:|
Tribst JM, Dal Piva AM, de Lima DR, Borges AL, Bottino MA. Simulated damage of two implant debridement methods: Nonsurgical approach with Teflon and stainless steel hand scalers. J Indian Soc Periodontol 2018;22:340-4
|How to cite this URL:|
Tribst JM, Dal Piva AM, de Lima DR, Borges AL, Bottino MA. Simulated damage of two implant debridement methods: Nonsurgical approach with Teflon and stainless steel hand scalers. J Indian Soc Periodontol [serial online] 2018 [cited 2020 Feb 25];22:340-4. Available from: http://www.jisponline.com/text.asp?2018/22/4/340/237197
| Introduction|| |
The oral cavity is colonized by approximately 750 species of microorganisms that form dental biofilm when organized and involved in an extracellular polysaccharide matrix., The biofilm can adhere to soft, solid, natural, or artificial surfaces of the oral cavity  such as dental implants, leading to pathologies relevant to dentistry. Among oral pathologies, dental caries and periodontitis are the two most common diseases related to dental biofilm action.
In patients with dental implants, biofilm accumulation in the peri-implant region may develop an inflammation reaction called peri-implantitis, which is similar to periodontitis. This inflammatory process promotes progressive bone loss, thus exposing more threads and allowing more bacterial proliferation. Exposed threads in contact with the oral environment provide a favorable niche to bacterial adherence.
Mechanical scaling with periodontal hand scalers and ultrasonic tips is the most common treatment of periodontal and peri-implant tissue infections.,, Periodic scaling and cleaning of biofilm accumulation niches are a noninvasive option, acting in prevention and production of excellent clinical results.,,, In general, the combination of this control therapy associated with instruction and patient hygiene motivation culminates in preventing the disease progression., Despite the similarities, implant prostheses need extra care, since titanium surface damages could facilitate the accumulation of bacterial biofilms ,, due to the increase in surface roughness.
Some studies suggest the use of plastic tips for handscalers and ultrasonic instruments for plaque removal due to their less structural damage in implants and components in comparison to metallic instruments.,,,, Using scanning electron microscopy it has been reported that the use of metal hand scalers is capable of damaging the titanium surface , so that results of surface roughness by optical profilometry demonstrate greater depth and width of the defects caused by the instrumentation performed with these hand scalers compared to plastic hand scalers. In spite of this, there are no reports in the literature of the residual stresses generated between the friction of the implant surface and the different instrument tips, which could aid in visualizing mechanical problems and in preventing plastic deformations.
Among the methods used in implantology,, computational mathematical analysis using the finite-element method is widely used which can demonstrate the effect of contact between materials. Thus, the purpose of this study was to compare the effect of mechanical prophylactic therapy on residual stresses on the implant surface and on the active tip of the instrument, varying the type of hand scaler material. The hypotheses of the study were that: (1) the rigid hand scaler would concentrate more stress in the implant and (2) there would be no difference in the stress concentration in the hand scalers.
| Materials and Methods|| |
For this study, a previously validated bone-level tapered implant model (Straumann Institute, Waldenburg, Switzerland) of 4.1 mm × 10 mm was used. The complete three-dimensional (3D) model presented a cemented fixed prosthesis on solid abutment and the implant was inserted into a slice of cortical and medullary bones. Each geometric structure of the components of this system was represented, including a fixing screw and cement line between the crown and abutment. Following ISO 14801, the implemented 3D model presented 3 mm of the exposed threads representing a bone defect of 4 walls with insertion loss. For simulation of a prophylactic mechanical debridement, a generic model of periodontal hand scaler was created based on the shank geometry of Gracey's hand scaler. The active face of the shank was disposed of in contact with the last thread exposed at a 90° angle. Each solid was considered a 3D volumetric body and exported in STandard for the Exchange of Product model data format for the analysis software.
In computer-aided engineering software (ANSYS 17.2, ANSYS Inc., Houston, TX, USA), the 3D model was subdivided into a finite mesh of quadratic tetrahedral elements totaling 651,239 nodes and 390,818 elements. A mesh convergence test with 10% coherence was used for this. Next, the contacts between the bodies were defined as being bonded to all components of the implant prosthesis. The type of contact selected for the hand scaler and implant was rough. The system fixation occurred at the cortical bone base and the applied force was in the Z (occlusal) direction with an intensity of 10 N in the hand scaler cable. Displacement of the hand scaler in the X and Y directions was prevented from occurring by lateral movements between the hand scaler and the implant [Figure 1]. The structural information required to perform the mechanical structural analysis was the elastic modulus and Poisson ratio, as based on previous studies [Table 1].,,,,, Two different simulations were performed: the hand scaler (1) presenting the mechanical properties of stainless steel or (2) Teflon.
|Figure 1: Boundary conditions: Load applied to pull the hand scaler in an occlusal direction with 10 N|
Click here to view
|Table 1: Mechanical properties of each material and structure used in the study: Elastic modulus in gigapascal and Poisson ratio (adimensional)|
Click here to view
| Results|| |
The results required for the implants and for the shank of the hand scaler were in von-Mises stress (KPa). A qualitative stress map was photographed in these structures and the 500 largest stress peaks in these structures were exported and plotted in bar charts for qualitative comparison. In the implant body [Figure 2], it is possible to notice greater residual stress accumulation on the titanium surface which was debrided with a metal hand scaler. Thus, in the bar graph of [Figure 2], it is possible to notice a superior ability of the Teflon hand scaler to soften damage. Considering the hand scalers themselves, the most stressful region was the active portion of the shank, highlighted with circles in [Figure 3]. A great similarity can be seen between the stress distribution of both instruments, as well as the magnitude of the stress peaks [Figure 3].
|Figure 2: Residual stress on the titanium surface according to the hand scaler material used in the computational test load application: (a) Teflon; (b) Steel|
Click here to view
|Figure 3: Residual stress in the hand scaler according to the material type: (a) Teflon; (b) Steel|
Click here to view
| Discussion|| |
The purpose of this study was to compare the effect of mechanical prophylactic therapy on residual stress on the implant surface and on the active tip of the hand scaler, varying the type of hand scaler material. The results of the present study demonstrate that the residual stress generated by the debridement can be modified depending on the instrument material, thus accepting the first hypothesis.
[Figure 3] shows that the implant surface can be impacted by more stress from the stainless steel hand scaler, justifying why some studies have found greater roughness and grooves in the titanium when this instrument is used. Von-Mises stress demonstrates possible failure regions in metal structures through finite-element analysis, and it can be extrapolated as a predictor that the more stressed regions will fail earlier. Despite this, the results of the present study serve to illustrate how two-hand scalers indicated for the same treatment and with the same contact force can induce different results in the implant. As the literature already reports the full capacity for prophylaxis with plastic hand scalers,,,,,, it is clear that its indication is less harmful for use on delicate surfaces. However, the present study did not simulate a series of oscillatory movements, such as what was promoted by using an ultrasonic device. The literature reports that ultrasonic prophylaxis is more efficient and practical by removing biofilm more easily in a shorter clinical time., As the ultrasonic tips can be found in plastic and metal, the results of the present study can serve as a theoretical basis for selecting plastic tips, due to its less aggressive behavior on the implant. In the same way, metal instruments and burs are only recommended in cases requiring the smoothening of surface roughness.
Several studies have been conducted with the purpose of analyzing the effect of different prophylactic treatments on titanium implant surfaces.,,, The surface roughness method stands out as predominant, which, in association with photomicrography, show the different potential of damage in the implant structure. Thus, the present study is notable for bringing information regarding superficial residual stresses in both implant and periodontal instruments.
An important point of the results generated in the present simulation is the analysis of the hand scaler active tip. In [Figure 3], both tips of the instruments presented similar stress distribution. Thus, instrument geometry was more important than the structural property reported for the final result. Therefore, a plastic hand scaler under the same conditions will generate as much stress as a metal hand scaler. In this way, plastic hand scalers require more attention from the clinician because, as the resistance of the plastic is inferior, failures such as shape loss and fractures can occur more easily by modifying the active tip of the instrument. Therefore, the second hypothesis was also accepted.
Depending on the type of device used and the prophylactic method, changes can be made on the surface of the implant,,,, especially when using a metallic instrument., This assertion is in accordance with the findings of the present study, which demonstrate an increase in the noxious stresses for this therapeutic modality when using metal hand scalers. Although the oral environment presents a variety of anatomy in each clinical case due to the pathogenicity of the microbiota and host response, the bone defect herein was created to allow the hand scaler complete freedom of movement. This should be taken into consideration when choosing the prophylactic method to be used, since anatomical modifications may hinder the treatment. In addition, the materials simulated herein are isotropic and homogeneous, without defects or considering ideal contacts. Therefore, this study serves as a complement to the in vitro and in vivo studies in the aspect of elucidating the mechanical behavior generated by the friction between instrument and osseointegrated implant.
| Conclusions|| |
Among the limitations of this computational study, it can be concluded that the residual stresses during mechanical prophylactic therapy in the implant and in the hand scaler vary according to the type of instrument material. Moreover, the hand scaler in Teflon is less damaging to the implant, yet more susceptible to deformations and possible early failures.
Financial support and sponsorship
Conflicts of interest
There are no conflicts of interest.
| References|| |
Pereira CA, Romeiro RL, Costa AC, Machado AK, Junqueira JC, Jorge AO, et al.
Susceptibility of Candida albicans
, Staphylococcus aureus
, and Streptococcus mutans
biofilms to photodynamic inactivation: An in vitro
study. Lasers Med Sci 2011;26:341-8.
Zhou P, Liu J, Merritt J, Qi F. A yadA-like autotransporter, hag1 in Veillonella atypica
is a multivalent hemagglutinin involved in adherence to oral streptococci, Porphyromonas gingivalis
, and human oral buccal cells. Mol Oral Microbiol 2015;30:269-79.
de Oliveira JR, de Aguiar Almeida RB, das Graças Figueiredo Vilela P, de Oliveira FE, da Rocha RF, Jorge AO, et al.
Control of microorganisms of oral health interest with Arctium lappa
L. (burdock) extract non-cytotoxic to cell culture of macrophages (RAW 264.7). Arch Oral Biol 2014;59:808-14.
Shemesh M, Tam A, Aharoni R, Steinberg D. Genetic adaptation of Streptococcus mutans
during biofilm formation on different types of surfaces. BMC Microbiol 2010;10:51.
Jakubovics NS, Yassin SA, Rickard AH. Community interactions of oral streptococci. Adv Appl Microbiol 2014;87:43-110.
Louropoulou A, Slot DE, Van der Weijden FA. Titanium surface alterations following the use of different mechanical instruments: A systematic review. Clin Oral Implants Res 2012;23:643-58.
Matarasso S, Quaremba G, Coraggio F, Vaia E, Cafiero C, Lang NP, et al.
Maintenance of implants: An in vitro
study of titanium implant surface modifications subsequent to the application of different prophylaxis procedures. Clin Oral Implants Res 1996;7:64-72.
Lindhe J, Meyle J; Group D of European Workshop on Periodontology. Peri-implant diseases: Consensus report of the sixth European workshop on periodontology. J Clin Periodontol 2008;35:282-5.
Stacchi C, Berton F, Perinetti G, Frassetto A, Lombardi T, Khoury A, et al.
Risk factors for peri-implantitis: Effect of history of periodontal disease and smoking habits. A systematic review and meta-analysis. J Oral Maxillofac Res 2016;7:e3.
Koldsland OC, Wohlfahrt JC, Aass AM. Surgical treatment of peri-implantitis: Prognostic indicators of short-term results. J Clin Periodontol 2018;45:100-13.
Hakki SS, Tatar G, Dundar N, Demiralp B. The effect of different cleaning methods on the surface and temperature of failed titanium implants: An in vitro
study. Lasers Med Sci 2017;32:563-71.
Schmidt KE, Auschill TM, Heumann C, Frankenberger R, Eick S, Sculean A, et al.
Influence of different instrumentation modalities on the surface characteristics and biofilm formation on dental implant neck, in vitro
. Clin Oral Implants Res 2017;28:483-90.
Renvert S, Polyzois IN. Clinical approaches to treat peri-implant mucositis and peri-implantitis. Periodontol 2000 2015;68:369-404.
Kourkouta S. Peri-implantitis management in the esthetic zone in a periodontally compromised patient: Five-year results including cone beam computed tomography. Int J Periodontics Restorative Dent 2018;38:e8-16.
Cafiero C, Aglietta M, Iorio-Siciliano V, Salvi GE, Blasi A, Matarasso S, et al.
Implant surface roughness alterations induced by different prophylactic procedures: An in vitro
study. Clin Oral Implants Res 2017;28:e16-20.
Bertoldi C, Lusuardi D, Battarra F, Sassatelli P, Spinato S, Zaffe D, et al.
The maintenance of inserted titanium implants:In vitro
evaluation of exposed surfaces cleaned with three different instruments. Clin Oral Implants Res 2017;28:57-63.
Ronay V, Merlini A, Attin T, Schmidlin PR, Sahrmann P.In vitro
cleaning potential of three implant debridement methods. Simulation of the non-surgical approach. Clin Oral Implants Res 2017;28:151-5.
Tribst JP, Dal Piva AM, Borges AL. Biomechanical tools to study dental implants: A literature review. Braz Dent Sci 2016;19:1-11.
Tribst JP, Rodrigues VA, Borges AL, Lima DR, Nishioka RS. Validation of a simplified implant-retained cantilever fixed prosthesis. Implant Dent 2018;27:49-55.
Tribst JP, Dal Piva AM, Shibli JA, Borges AL, Tango RN. Influence of implantoplasty on stress distribution of exposed implants at different bone insertion levels. Braz Oral Res 2017;31:e96.
Tribst JP, de Morais DC, Alonso AA, Piva AM, Borges AL. Comparative three-dimensional finite element analysis of implant-supported fixed complete arch mandibular prostheses in two materials. J Indian Prosthodont Soc 2017;17:255-60.
] [Full text]
Danza M, Palmieri A, Farinella F, Brunelli G, Carinci F, Girardi A, et al.
Three dimensional finite element analysis to detect stress distribution in spiral implants and surrounding bone. Dent Res J (Isfahan) 2009;6:59-64.
Tian Y, Mu Y, Setzer FC, Lu H, Qu T, Yu Q, et al.
Failure of fiber posts after cementation with different adhesives with or without silanization investigated by pullout tests and scanning electron microscopy. J Endod 2012;38:1279-82.
Zarone F, Sorrentino R, Apicella D, Valentino B, Ferrari M, Aversa R, et al.
Evaluation of the biomechanical behavior of maxillary central incisors restored by means of endocrowns compared to a natural tooth: A 3D static linear finite elements analysis. Dent Mater 2006;22:1035-44.
Madfa AA, Kadir MR, Kashani J, Saidin S, Sulaiman E, Marhazlinda J, et al.
Stress distributions in maxillary central incisors restored with various types of post materials and designs. Med Eng Phys 2014;36:962-7.
Ge C, Dong Y, Maimaitituersun W. Microscale simulation on mechanical properties of al/PTFE composite based on real microstructures. Materials (Basel) 2016;9. pii: E590.
Deǧer Y, Adigüzel Ö, Yiǧit Özer S, Kaya S, Seyfioǧlu Polat Z, Bozyel B, et al.
Evaluation of temperature and stress distribution on 2 different post systems using 3-dimensional finite element analysis. Med Sci Monit 2015;21:3716-172.
Schmage P, Kahili F, Nergiz I, Scorziello TM, Platzer U, Pfeiffer P, et al.
Cleaning effectiveness of implant prophylaxis instruments. Int J Oral Maxillofac Implants 2014;29:331-7.
[Figure 1], [Figure 2], [Figure 3]